HD-B604B-S Computerized Servo Double Column Tensile Test Machine
it is suitable for testing and analyzing the mechanical properties for plastic film, composite film, flexible packaging material, adhesive tape, rubber, paper and other products in tensile, peeling, tearing, heat sealing, adhesion, compression, bending. And it is also suitable for testing and analyzing the mechanical properties for corrugated cartons, honeycomb cartons, flexible packaging bags, plastic drums (edible oil, mineral water), paper drums, paper cans, IBC drums and hollow containers.
|Load cell brand||Germany brand TECSIS load cell|
|Motor||400w Panasonic serve motor|
|Driver||400wPanasonic serve driver|
|Screw||High precision imported ball screw|
|Control Mode||Computerized ,windows mode operation|
|Test Speed||0.1-500mm/min (Adjustable)|
|Vertical space||1200mm（including clamp）|
|Power||1∮ AC 220V|
A tester integrates multiple independent testing functions such as tensile, deformation, peeling, tearing, and compression, and provides a variety of test items for users to choose;
● Provide data analysis of constant tensile stress, elastic modulus, stress and strain;
● Double-pillar ball nut and screw drive to ensure high precision and smooth operation;
● The equipment adopts miniaturization and structural integration design, which can meet the test requirements of various occasions;
● 1~500mm/min stepless speed change, which provides convenience for users to test under different test conditions;
●Intelligent configuration such as multi-level limit protection, overload protection, automatic high-speed return, and fault notification to ensure the user's operation safety;
● Embedded system and membrane key switch, which are convenient for users to conduct test operations quickly;
● The embedded control system effectively guarantees the security of the system and improves the reliability of data management and test operations;
● The system is equipped with 4 USB ports and 2 network ports to facilitate data transmission and external access of the system;
● Professional software provides statistical analysis of group test curve overlay analysis, maximum, minimum, average, and standard deviation.
HD-B604B-S has rich applications, equipped with more than 100 different sample fixtures for users to choose, can meet the test requirements of more than 1000 kinds of materials; According to the different materials of users, haida also provides customized services to meet the testing needs of different users.
SOME APPLICATION EXAMPLES
|Basic application||Extended application (Need special accessories or restructuring)|
|Tensile strength and deformation rate||Breaking force of ampoule||Film puncture force||Puncture force of lid of infusion bag with lid||Puncture/pulling force of soft rubber stopper|
|Breaking force||Combination cover opening force||ZD bottle cap tearing force||Oral fluid lid tearing force||Oral fluid cap puncture/pulling force|
|Heat seal strength||Pulling force of infusion bag cover inclined at 90 degrees||Pull-out force of lid for infusion bag with lid||23 degree tilting cap pulling force||Piercing/pulling force with bottle cap and stopper|
|Tear resistance||90 degree peel force of tape||Perfect bound pages tearing force||90 degree water-based plaster peeling force||Adhesive tearing force|
|180 degree peel||Adhesive strength test (soft)||Adhesive strength test (hard)||Peeling force of hose cap||Catheter and catheter joint disengagement force|
|90 degree peel||Makeup brush bristles pulling force||Toothbrush bristles pulling force||Rope breaking force||Jelly cup and yogurt cup opening force|
|Fixed tensile test||Milk cup cup film peeling force||Extraction force of rubber plug||45 degree peeling force of bottle film||Ziplock bag mouth pullforce|
|Down test||Magnetic card core peeling force||90 degree peeling force of magnetic card||Heat-sealing film tearing force||Protective film separation force|
|Puncture retention||Release paper separation force||Trouser tearing force||Unwinding force of tape||Pressure resistance of plastic bottles|
SOFTWARE – TM 2101
Curves display picture
Estimated Packaging Dimensions: (W*D*H) 690*1080*1790mm
Power supply source: single-phase, 220V±10%, 50/60Hz (can be appointed)
Estimated Gross Weight: 270kg
HS Code: 9024800000
TENSILE CLAMP LIST FOR OPTION
HAIDA TEST EQUIPMENT TEST STANDARDS FOR SHARING
|Adhesives Tape/Pressure Sensitive TapeØ EN 1719: Loop Tack TestØ ASTM D3654: Shear Adhesion Test Ø ASTM D3759: Tensile &Elongation TestØ FTM: 180&90 Degree Peel Test, Loop TackØ EN 1939, ASTM D3330: 180&90 Degree Peel TestØ ISO 11339, ASTM D1876, ASTM F2256: T-type Peel Test|
|AdhesiveØ ASTM D903: Peel & Stripping Test Ø ASTM D1002: Shear Strength (Adhesive for metal to metal)Ø ISO 4587: Tensile lap-shear strength (Rigid-to-rigid bonded assemblies)|
|Film/Plastic FilmØ ASTM D882: Tensile TestØ ASTM F1306: Puncture TestØ ASTM D1938: Trouser Tear TestØ ASTM D2732: Thermal Shrinkage TestØ ASTM D1709, ISO 7765-1: Free-Falling Dart MethodØ ASTM D1894: Static and Kinetic Coefficients of FrictionØ ASTM D1424, ASTM D1922, ISO 1974: Elmendorf Tearing Strength Test|
|Plastic& Plastic PipeØ ISO 180: Izod Impact TestØ ISO 9966: Ring Stiffness TestØ ISO 179: Charpy Impact TestØ ISO 4892: Weathering Aging TestØ ASTM D3163: Plastic Lap-Shear TestØ ISO 4422, ISO 3127: Pipe Impact TestØ ASTM D4565, ISO 11357, EN 728: DSC/OITØ ISO 75: Heat Deflection Temperature (HDT)Ø ASTM D1598: Hydrostatic Pressure and Burst Test Ø ISO 1133, ASTM D1238: Melt Flow Rate Test (MFR/MVR)Ø ASTM D790, ISO 178, ISO 14125, ASTM D6272: Bend TestØ ISO 306, ASTM D1525: Vicat Softening Temperature (VST)Ø ASTM D882, ASTM D 638, ISO 527, ASTM D1708: Tensile Test|
|RubberØ ASTM D412, ISO 37-2005, JIS K6251, DIN 53504-2009: Tensile Strength & Elongation TestØ ASTM D1149-07, ASTM D1171-99, ISO 7326, ISO1431-1, DIN 53509-1: Ozone Aging TestØ ASTM D624-00, ISO 34-1: Tearing strength TestØ ASTM D395-03, ISO 815-1: Compression Set TestØ ASTM D2240-05, ISO 7619-1, JIS K6253, ISO 868: Shore Hardness Type A/D Ø ASTM D573, ASTM D865, ISO188 Method B, DIN 53508 High/ Low Temperature TestØ ISO 2781(Method A): Density TestØ ASTM D5289-95/ ISO6502: No-Rotor Rheometer/CuremeterØ ASTM D1646: Mooney Viscometer|
|FoamØ ASTM D3574 Test E, ASTM D1623, ISO 1798: Tensile TestØ ASTM D3574 Test F, ISO 8067: Tear TestØ ASTM D3574 Test A: Density TestØ ASTM D3574 Test B1, ISO 2439: IFD TestØ ASTM D3574 Test H: Resilience (Ball Rebound) TestØ ASTM D3574 Test I3: Dynamic Fatigue Test by Constant Force|
|MetalØ ASTM E8, EN 10002-1,ASTM A370: Tensile TestØ ASTM A370: Bend TestØ ASTM B-117, ISO 3768: Salt Spray TestØ ISO 6508: Rockwell Hardness TestØ ISO 6507: Vickers Hardness TestØ ISO 6506, ASTM E10-14: Brinell HardnessØ ASTM E23, ISO 148: Impact Test|
1. The guarantee period with FOC within 1 year (not including expendable parts cost and transport & travel fees)
1.1 During the guarantee period, Haida shall supply free maintenance or replacement for the damaged part (just for non-expendable part) caused by non-human reasons;
1.2 If any quality problems occur within the guarantee period, and Haida must provide on-site service, the transport and travel expense shall be borne by buyer;
1.3 If any big quality problems occurs out of the guarantee period, Haida will provide a maintain service, the transport and travel expense shall be borne by buyer, also charge for a favorable price for new parts;
1.4 Haida will provide a lifetime favorable price to the buyer for expendable & non-expendable parts used in system operation, equipment maintenance;
2. The following conditions need to be paid reasonably even in the guarantee period:
2.1 Natural disaster
2.2 Operating mistakes
2.3 Voltage is not fit for our manual
2.4 Uninstall without our guides
2.5 Damaged for borrowing to others
2.6 Damaged for without authorized machine modification
2.7 Damaged for without authorized calibration
2.8 Without authorized transshipment mistake
3. Attentions: the machine cannot be used following situations:
3.1 Vibration, rocking place.
3.2 Direct sunlight.
3.3 Hot, dusty, damp places.
3.4 To ensure safe, AC supply of the machine should be well grounded.
3.5 Do not use strong solvents (such as: benzene, nitro oil) washing machine.
3.6 Do not inject water and debris into the machine to prevent damage to electrical components and electrical shock.
3.7 Machine’s disassembly and debugging can only be measured by the State Department approved the units and the company, other people not allowed to overhaul.
PART OF OUR COOPERATED PART