HD-F731 Chair Swivel Tester
HD-F731 Chair Swivel Test Rig
only for reference
Two loading pad on Chair Electromotive test axis for the rotation of the turning Operation interface
Test rig for rotation testing on seating furniture according to BIFMA X5.1, load controlled with loading pad. The pneumatic test axis operates from PLC.
Machine Dimensions: (W x D x H) 1680*890*1840mm
Power supply source: single-phase, AC220V±10%, 50/60Hz (can be appointed)
Gross Weight: 400kg
1 framework with traverse member and base plat 1200mm ×1200 mm , 12mm thick, galvanized steel
Rotation test axis
1 electromotive test axis for the rotation of the turning knob, position-controlled, rotation angle +/- 360°
Pneumatic test axis
1 pneumatic test axis for seat, load controlled, position diameter 80mm, stroke 500mm
Load Cell 122 kg (270 lb.) . Load is transmitted vertically to the seat with lateral adjusting range
Operating pressure and cylinder limit switch control
1 emergency button
Number and Cycles
Can be set by PLC ( 0- 99,999)
1 load pad diameter 200mm DIN EN 1335
Reference Doc: BIFMA X5.1-2017 Swivel Test
Chair Test Step
a) The chair shall be restrained on a platform. Either the seat or the platform shall be restrained from rotation.
b) If the seat adjustable, set it to the maximum seat position. Set all other adjustable features to the normal use condition.
c) A 122 kg (270 lb.) load shall be placed on the seat such that the centre of gravity of the load is 51 to 64 mm (2 to 2.5 in) forward of the centerline of the rotational axis as shown in Picture 1.
d) The cycling device shall be adjusted to rotate the lesser of the following: the available range of rotation or 360 degrees ± 10 degrees. If the available range of rotation is less than 360 degrees, the rotation of the test machine shall be adjusted such that the swivel mechanism touches but does not override the stops. The rotation may be either bidirectional or unidirectional.
e) For chairs that swivel 360 degrees, a cycle is one full rotation. For chairs that swivel less than 360 degrees, one cycle is rotating from one stop to the other stop.
f) The seat or platform shall rotate for 60,000 cycles at a rate between 5 and 15 rotations per minute.
g) If the seat height is adjustable set the height to its lowest position.
h) For all chairs, continue the test for an additional 60,000 cycles to a total of 120,000 cycles.
A) Quality assurance items
The guarantee period with FOC within one year (not including nonexpendable & transport and travel fees)
1. During the guarantee period, Haida shall supply free maintenance or replacement for the damaged part(just for nonexpendable part) induced by non-human damage;
2. If any quality problems occurs within the guarantee period, and Haida must provide on-site service, the transport and travel expense shall be borne by customer;
3. If any big quality problems occurs out of the guarantee period, Haida will provide a maintain service, the transport and travel expense shall be borne by customer, also charge for a favorable price;
4. Haida will provide a lifetime favorable price to the buyer with the materials and spare parts used in system operation, equipment maintenance;
B) The main quality warranty maintenance certificate
If there is any dispute please refer to our guarantee letter, so:
① please keep the guarantee letter, if you lost it, please connect with us in a month.
② If the guarantee letter has been altered or it has no our stamp, it is useless.
C) The following conditions need to be paid reasonably even in the assurance period:
① Natural reasons
② Operating mistakes
③ Voltage is not fit for our operation instruction
④ Repack it without our guides
⑤ Damaged for borrowing to others
⑥ Damaged for authorized machine modification
⑦ Damaged for authorized calibration
⑧ Authorized transshipment mistake
⑨ Serve for long distance area
① Any service outside Guangdong, China, transport and travel fees shall be paid by customer.
② The apparatus try not to be used in the following situations:
a. Vibration, rocking the occasion.
b. Direct sunlight.
c. Hot, dusty, damp places.
d. To ensure safe, AC supply of the machine should be well grounded.
e. Do not use strong solvents (such as: benzene, nitro oil) washing machine.
f. Do not inject water and debris into the machine; prevent damage to electrical components and electrical shock.
g. Instrument displays the disassembly and debugging can only be measured by the State Department approved the units and the company, other people not allowed to overhaul.